Surgical Guide Resin(#3165)
- Biocompatible resin
- Easy to peel off the support by hand
- Low shrinkage, stable storage, no deformation and shrinkage after curing
- High hardness with some toughness
- Disinfected with iodine volts, no irritating odor
- UV curing wavelength: 405nm
Features of Surgical Guide Resin(3165)
It is an excellent resin material for 3D printing surgical guide.
Safe for surgical guide
The surgical guide resin is a biocompatible resin for applications such as 3D printing dental surgical guides for implants. It has undergone rigorous testing and fully meets the dental requirements for part quality, precision and performance.
Stable and not deformed
Combine strength and toughness. Strength can withstand force without deforming, while toughness can withstand force without fracturing.
Precise fit to the teeth
The guide plate printed with 3165 resin has accurate hole position, high hardness and toughness; it can easily install the guide ring and withstand the mechanical impact of surgery.
Low water absorption
Water absorption rate is less than 0.5%. For post-sterilization, the surgical guide is soaked in water containing iodophor for 30 minutes and will not absorb water and become soft or damaged.
How to post-process
Proper post-processing is very important. You can refer to our video to do it. If you have any questions, please feel free to communicate with us.watch video
Specifications of Surgical Guide Resin(3165)
Content Data value Testing method Viscosity 330-350cps(25℃) ASTM D1084-1997 Density 1.15 ASM D1875-69(1980) Shrinkage ＜0.5% Capillary method Hardness 78D ASTM D2240-05(2010) Water absorption 0.50% ASTM D570 Bending strength 65MPa ASTM D1781 Transmittance 80%”}”>>80% ASTM D747 Biocompatibility TEST PASS ISO10993-1/ISO10993-10
You can download the parameter here.
Part removal, rinsing, drying, post-curing, support removal, and polishing.
1. Post-curing time and temperature can lead to suboptimal mechanical and biocompatibility. Please find the right curing time.
2.Disinfection method for the guide plate: Firstly, disinfection by soaking in iodophor solution for 30 minutes, then deiodination by rinsing with medical alcohol or physiological saline.
3.In the data layout, the hole position of the guide ring is kept as vertical as possible to ensure the accuracy of the size.
4.If the hole collapses, It can be reamed by using a drill bit later.
Request a quote?
We’re here to help you find the best, most cost‑effective solution for you.